For upgrade options, visit our 'Which Meter is right for you? Accu-Chek Aviva test strips help make every strip count, by making it easy to test right the first time. In addition to fingertip testing, the Accu-Chek Aviva Nano blood glucose monitoring system can be used for testing on alternate sites, including the palm, forearm, upper arm, thigh or calf. Alternate site testing requires the use of a special lancet cap.
|Published (Last):||28 January 2010|
|PDF File Size:||20.37 Mb|
|ePub File Size:||8.76 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Table of Contents. Troubleshooting troubleshooting checks 26 Chapter 5: Maintenance and Troubleshooting 60 maintenance and troubleshooting 63 screen messages and troubleshooting Suitable for self-testing. Page 4 The meter is for outside the body in vitro use.
It should not be used to diagnose diabetes. Other strips will give inaccurate results. You may need to change the time to your time zone. Page 9 Golden End — Insert this end of the strip into the meter. Yellow Window — Touch blood drop or control solution here. Make sure the meter 2. Turn the meter over. Remove the old code is off. Change the code chip every time you open a new box of strips! Page 11 4. Turn the code chip over so the 5. Leave the code chip in the meter code number faces away from until you open a new box of you.
Push it into the code chip strips. You need the meter, a strip, and a lancing device with a lancet loaded. Wash and dry your hands. Page 15 Yellow Release Button 5.
A strip and 6. Lance your fingertip with the lancing device. Blood samples flashing blood drop taken from the palm are equivalent to blood taken from the symbol appear on fingertip. For detailed information on how to obtain blood the display. Page 16 7. Gently squeeze your finger to assist 8. Touch the blood drop to the front edge the flow of blood. This helps you get a of the yellow window of the strip. Do not put blood on top of the strip.
Page 17 9. The result appears on the display. If you want to mark the test result as a pre-meal or post-meal test, or with a general marker, leave the strip in the meter see the next section.
Otherwise, remove and discard the used strip. Note: After a successful test, the meter turns off five seconds after the strip is removed. Page 19 Here is how to mark a test result and initiate a post-meal reminder: Perform a blood glucose test. Do not remove the strip. Press and release to toggle through the test result markers and post-meal reminder. Blood obtained from a fingertip or palm can be used at any time to measure blood glucose.
The two palm testing sites are the fleshy areas under the thumb thenar and under the little finger hypothenar. Page 22 To perform a blood glucose test from an alternate site, you need the meter, a strip, and a lancing device with the clear AST cap.
Prepare the lancing device. Make sure the code number on the display matches the code number on 2. Page 23 4. A strip and flashing 5. Press the lancing device 6. Trigger the lancing blood drop symbol firmly against a fleshy device while keeping appear on the display.
Press the lancing site. Page 24 7. Touch the blood drop to the front 8. When you see flash, you have the used strip. Note: If the blood drop is too small, reapply pressure to get a sufficient blood drop. Check if the strips are expired. Replace the strips if you think they 2. Page Troubleshooting Checks 5. Check if you followed the testing steps. If you still have problems, call Accu-Chek Customer Care at Check if the code number on the meter If they do not match, insert the correct display matches the code number on code chip into the meter and test again.
Here are the most common symptoms: High blood glucose hyperglycemia : fatigue, increased appetite or thirst, frequent urination, blurred vision, headache, or general aching. You can review them at any time. Test results are stored from the newest to the oldest. Page Viewing Test Results 7, 14, 30, and 90 day averages. To view stored control results, first download them to a compatible software application. For product availability, please contact Accu-Chek Customer Care at Viewing Test Results With the meter on or off, press and release to enter memory.
Page 30 Note: Only test results that have been assigned a pre-meal or post-meal marker are included in pre-meal and post-meal averages. All blood glucose results are included in the general 7, 14, 30, and 90 day averages.
Past Results General Average Press Here are the features you can customize in the order in which they appear in the set-up mode: Time and date — set the time and date Beeper — Press and release 2. Page 33 decrease increase 4. Press and release to confirm the 5. Press and release to adjust hour. The minutes flash. Press and release confirm the minutes. Page 34 6. If you want to set up more options, day and year. If you want to exit, press and hold until you see the flashing strip symbol.
Page 36 1. Press and hold until 3. Press and release to turn the meter on. Page 37 4. Press and release 5. Page 39 1. Page 40 4. Page 44 4. Press and release to switch hour flashes. Page 45 7. Press 8. Press and release 9. These are the only on the display. Page 47 2. Press and release 4. Page 48 5. The display shows 3. Press and release to select the level you want. Press and release confirm it.
Press and hold until you see the flashing strip symbol to exit the set-up mode. Infrared IR Window 1.
Accu-Chek Aviva Nano product support
Its small, sleek design fits in the palm of your hand, and the brilliant backlit display makes reading your numbers easy—any time, anywhere. A battery icon means the battery power is low. Change the batteries soon. Download the owner's booklet now and refer to Chapter 7, "Changing the Batteries.
Accu-Chek Performa Nano product support
Watch video. Make sure you have two new 3-volt, lithium batteries, number CR, to insert. Open the battery door on the back of the meter, remove the old batteries for at least 20 seconds, and press any button on the meter. Snap the battery door back into place. Setup and hour are flashing on the display. The meter uses 2 3-volt lithium coin cell batteries type CR found in many stores. The meter includes an additional internal battery that retains the time and date while you change batteries.
Accu-Chek Aviva Nano
With its discreet design and smart simplicity, the Accu-Chek Performa Nano system offers style and performance in a reliable small meter. Accu-Chek Performa test strips help make every strip count, by making it easy to test right the first time. For further information on how to use your meter, please see the Accu-Chek Performa Nano reference manual. Only 3 other substances in the blood stream galactose, ascorbic acid, and lipids can interfere with the Accu-Chek Performa test strips. Control test values do not stay in the memory that can be accessed on the meter.